Involvement:Expert (understands the inner workings) - Ownership:more than 1 month
It's really cheap. English manual and batteries included. Probes are reasonable for such price. Overall case built quality is good. Case is sturdy and doesn't feel cheap. LCD is bright and big enough. There is a HOLD button, witch is really handy. Resistance matter is really accurate; it shows the same as my Fluke does in same mattering ranges.
Accuracy is pretty horrible. You can not trust this clam matter at all, even measuring voltages. On AC scale you usuallly get +/- 4V error. On DC scale - +/- 2V error. Current matter shows nonsense really. When measuring mine flat power input, I get ~3A, even when washing machine (>1kW), TV, PC and electrical kettle (>2kW) is on. Live wire indicator ("LIGHT" LED) is useless too: it doesn't show witch power line is live, only shows that there are power in the line.
Matters casing is welded together, so you can not take it apart. This way internal construction is totally unknown. Maybe it could be re-calibrated or fixed for more accurate measurement.
Well, this matter is only suitable for inaccurate measurements, for example, if you just want to check if the is voltage and current in the power line or to measure power cable resistance. I you want to get accurate reading, you should by more expensive branded clam matter.
Note:Stock and Availability shown on this site is for your reference only. While
we strive to provide the most accurate and timely stock and availability information,
availability information may become out of date and may change between the time
you added an item to cart and the time your order is received.
Quantities on clearance items are limited. Prices are current at time of posting. DX Reserves the right to change prices at any time without notice.