Involvement:Expert (understands the inner workings) - Ownership:1 week to 1 month
REAL BANDWIDTH! In case 1:10 I measured bandwidth about 70MHz for -3dB (sinus generator TGR1040, reference Agilent probe 1:10@300MHz N2863B and scope DSO-X2022A) In case 1:1 you can count with about 35MHz bandwidth for -3dB.
For orientation measure not bad for this price, but for serious measure you should choose better probe. A have 20MHz analog scope - for this equipment is Ok, but for my 100MHz TEK i will look for somethig better.
Real bandwidth is much lower than is written and expected.
Note:Stock and Availability shown on this site is for your reference only. While
we strive to provide the most accurate and timely stock and availability information,
availability information may become out of date and may change between the time
you added an item to cart and the time your order is received.
Quantities on clearance items are limited. Prices are current at time of posting. DX Reserves the right to change prices at any time without notice.