Well manufactured, good packing, price is OK (in compare with other similar products). Version TX3125 has pretty good frequency characteristic - but.... (see cons). I was very unpleasant surprised with voltage derating curve versus measured frequency. E. g. only 800V maximal input voltage on 100 kHz frequency, 600V on 200 kHz etc. Too weak for 250 MHz ... Read more
1)Perfect for use with oscilloscopes Rigole, in my case, a model RIGOL DS1052E. 2)Excellent quality materials used. 3)Excellent equipment included in the package. 4)The set included clear and detailed instructions to use that helps to immediately deal with this product. Deficiencies in this product has been found that it is necessary! Fast delivery ... Read more
x100 with accuracy about 1%. Price lower then other x100 probes. Red color - for human safety. Datasheet included. Non-inductive adapter included. max compensated input capacistance = 23pF. This make probe unusable with many oscilloscopes, which input capacistance is 25pF and higher. Datasheet comes with this probe, says in specifications: "10-50pF ... Read more
On 100Mhz Rigol is shows rise-time 2.2ns compared to 2.7-2.8ns on bundled 60Mhz probes. Build quality is very good. Looks decent to work with 220V, I wouldn't do it with default Rigol probes. It is very expensive. Well, now when I see that it's not much better than 60/100Mhz probes on my scope. 1:100 with no switch might be a problem for small signals.... Read more
Note:Stock and Availability shown on this site is for your reference only. While
we strive to provide the most accurate and timely stock and availability information,
availability information may become out of date and may change between the time
you added an item to cart and the time your order is received.
Quantities on clearance items are limited. Prices are current at time of posting. DX Reserves the right to change prices at any time without notice.