Nice oscilloscope – DSO1060 5.7" LCD 60MHz 2-Channel Handheld Scopemeter Oscilloscope with USB Host/Device Reviews
posted by budavargas on 05/27/2011
Involvement:Expert (understands the inner workings) - Ownership:more than 1 month
At first sigth: A sturdy portatile oscilloscope with a big screen and a nice carring bag. It has an intuitive stable software with drivers included (only for Windows, sadly) and an engrish manual. Easy to operate after a while. Can take bmp screenshots and store them directly to an USB drive pluged.
Testing: I was able able to see signals up to 60 [MHz]. Also, I generated a 100 [nsec] pulse at 500[Hz] for transcient analysis and I got quite amazed. Finally, it has a good 50-60 [Hz] issolation (220 VAC/50 Hz in my country). For test results, see the photo section
Battery efectively last for 5 to 6 hours. I've been using this oscilloscope for the last 6 month and battery life is still the same.
I found that combining a multitester with an oscilloscope is a very clever idea. Very useful to carry a single instrument to do several measures. Measures using the multitester are stable and within my error tolerance.
It's most impresive fact it's is price. For 450 bucks, you have a very nice instrument ideal for a basic design and development bench.
FFT: Operations are basic. There is no options like center frequency, span, sweep time or bw settings expected on a Spectum Analizer or a more sophisticated oscilloscope
Probes: While tester probes are ok, scope probes are cheap. I recomend buying a couple at your local electronics shop or order them from a more respetable manufacturer. There are no fine probes here on DX
Screen: Not the best available, but sufficient. I got some random pixels lighted while using it.
Measures: While having a lot of posible automatic measures, many of them aren't as stable as I would like to. At high frequencies or low levels, they tend to jump from it's value to multiples (or fractions) of it.
60 [MHz] and 100 [nsec] pulses tests are shown on the photo section of this item.
-BK Presicion 2542 100[MHz] DSO
-HP 8594E Spectrum Analizer
-HP 8350A Sweep Oscilator
-Digilent Spartan 3E-1600 Development Board (for pulse generation)
An overall useful oscilloscope for hobby and semi-pro uses. It will not remplace a BK precition, Tectronic or any other well known manufacturer but it's nice to have some of their capabilities for 1/2 or 1/3 of their prices. Im glad with my purchase and will recomend it to every situation where a basic oscilloscope is needed.
Note:Stock and Availability shown on this site is for your reference only. While
we strive to provide the most accurate and timely stock and availability information,
availability information may become out of date and may change between the time
you added an item to cart and the time your order is received.
Quantities on clearance items are limited. Prices are current at time of posting. DX Reserves the right to change prices at any time without notice.